Usage: Metal or plastic spoons with or without holes can be used. To allow DentapressAPutty to adhere to the spoon, the spoon must be covered with a thin layer of Silicone-Adhesive. let Adhesive dry for 3 minutes. Take an equal amount of putty base and catalyst material with a special spoon (dark spoon is for the base, the light-colored spoon is for catalyst). Knead the two ingredients for at least 30 seconds until a homogeneous color is obtained. Place the kneaded material on the spoon and start measuring within 60 seconds after mixing starts. Wait for it to harden at least 2 minutes in the mouth.
Disinfection: The measure can be disinfected with normal commercial items. Please note that the disinfectant used is suitable for use in additional silicone materials. The use of 2% glutaraldehyde is recommended. The waiting time of the measure in the disinfectant is 10-15 minutes. is. Then wash for 15 seconds in running water. Store the measure at normal room temperature (15-25 °C) and prepare a plaster model within 30 minutes of oral administration.
Galvanoplasty: All of the measures can be plated with copper and silver.
Model preparation: The measure should be stored at normal room temperature. (max. 25 °C). The measure can be used for casting at least 30 minutes after oral administration. Apart from this, there is no limitation in terms of time.
Classification: DentapressAPutty complies with DIN ISO 4823 (2001) Type (0).
Storage: Do not store above 25 °C. Do not use it after the expiry date.
CAUTION: Do not mix the lid and measuring spoon. Do not use with standard measurement materials (C silicones). Do not use latex gloves, as latex may interact with polyvinyl on contact and distort the measurement.
Warranty: SeDenta Health Services Co. Ltd. warrants that this product is free from material and manufacturing defects. SeDenta Health Services Co. Ltd. does not take responsibility for the salability of the product or its suitability for a special purpose. Responsibility for proper and correct use of the product belongs to the user. Regarding any damage within the warranty period, SeDenta Health Services Co. Ltd.’s responsibility is only to repair or replace the relevant product.
Limitation of Liability: Regarding the use of this product, SeDenta Health Services Co. Ltd. is not responsible for any direct, indirect, special, or any kind of damages.
KEEP OUT OF THE REACH OF CHILDREN. FOR DENTIST USE ONLY.
It was produced in Germany for SeDenta Health Services Co. Ltd.
In 2017, according to its activities in Adana Region, our company produces Dentistry and Dental Laboratory Consumables and Medical products.