Indication: Single-phase measurement material for the measure of crowns, bridges, inlays, partial and complete prosthesis. Also used for feedstock measure, function measure and prosthesis repair, and copper ring measure.
Preparation of the cartridge: Before the cartridge cover is removed and the mixing tip is replaced, the mixing gun is squeezed to take out a quantity of measuring material in the size of chickpeas. The level of the base and catalyst measurement material at the end of the tube is equalized. Then the mixing tip is attached. Thus, the cartridge is ready for use. Additional “Intra Oral Tips” are foreseen for direct and precise application to the mouth. Do not cover the cartridge with its cap again. The used mixing tip acts as a cover until the next use. Otherwise, there may be a risk of contamination or undesirable hardening disorders.
Usage: The automatically mixed substance is applied to the putty material in the measuring spoon and directly to the mouth. The measuring spoon should be placed in the mouth in 1 minute 30 seconds and it should remain in the mouth for at least 2 minutes.
Disinfection: Make sure that the items are suitable for use. The use of 2% glutaraldehyde is recommended. The waiting time of the measure in the disinfectant is 10-15 minutes. Then wash it for 15 seconds in running water. Store the measure at normal room temperature (15-25 °C) and prepare a plaster model within 30 minutes after taking it out from the mouth.
Galvanoplasty: All of the measures can be plated with copper and silver.
Model preparation: The measurement material is compatible with commercially available plasters. Type 4 model plaster is recommended for model preparation.
Classification: DentapressA Medium conforming to DIN EN ISO 4823 (2001), Type 3 (fluid)
Storage: Do not store above 25 °C. Do not use it after the expiry date.
CAUTION: Do not mix the lid and measuring spoon. Do not use with standard measurement materials (C silicones).
Warranty: SeDenta Health Services Co. Ltd. warrants that this product is free from material and manufacturing defects. SeDenta Health Services Co. Ltd. does not take responsibility for the salability of the product or its suitability for a special purpose. Responsibility for proper and correct use of the product belongs to the user. Regarding any damage within the warranty period, SeDenta Health Services Co. Ltd.’s responsibility is only to repair or replace the relevant product.
Limitation of Liability: Regarding the use of this product, SeDenta Health Services Co. Ltd. is not responsible for any direct, indirect, special, or any kind of damages.
KEEP OUT OF THE REACH OF CHILDREN. FOR DENTIST USE ONLY.
It is produced by P.L. Superior Dental Materials GmbH, 22607 Hamburg Germany, for SeDenta Health Services Co. Ltd.
In 2017, according to its activities in Adana Region, our company produces Dentistry and Dental Laboratory Consumables and Medical products.