Indication: Crown, bridge, inlay, and onlay measurements. Function and feeding measures, Implant or transfer measures.
Preparation of the cartridge: Before the cartridge cover is removed and the mixing tip is replaced, the mixing gun is squeezed to take out a quantity of measuring material in the size of chickpeas. The level of the base and catalyst measurement material at the end of the tube is equalized. Then the mixing tip is attached. Thus, the cartridge is ready for use. Additional “Intra Oral Tips” are foreseen for direct and precise application to the mouth. Do not cover the cartridge with its cap again. The used mixing tip acts as a cover until the next use. Otherwise, there may be a risk of contamination or undesirable hardening disorders.
Usage: The automatically mixed substance is applied to the putty material in the measuring spoon and directly to the mouth. The measuring spoon should be placed in the mouth in 1 minute 30 seconds and it should remain in the mouth for at least 2 minutes.
Disinfection: The measure can be disinfected with normal commercial items. Please note that the disinfectant used is suitable for use in additional silicone materials. The use of 2% glutaraldehyde is recommended. The waiting time of the measure in the disinfectant is 10-15 minutes. is. Then wash for 15 seconds in running water. Store the measure at normal room temperature (15-25 °C) and prepare a plaster model within 30 minutes of oral administration.
Galvanoplasty: All of the measures can be plated with copper and silver.
Model preparation: The measure should be stored at normal room temperature. (max. 25 °C). The measure can be used for casting at least 30 minutes after oral administration. Apart from this, there is no limitation in terms of time.
CAUTION Do not use with condensation silicone or polyether measurement materials. Do not allow the measure to come into contact with organic solvents or substances containing them, as this may lead to deterioration or incompatibility of the measure. Some people may have an allergic reaction.
Storage: Do not store above 25 °C. Do not use it after the expiry date.
Classification: DentapressA Light is Type 3 (fluid), conforming to DIN EN ISO 4823 (2001). Type 3 (fluid)
Warranty: SeDenta Health Services Co. Ltd. warrants that this product is free from material and manufacturing defects. SeDenta Health Services Co. Ltd. does not take responsibility for the salability of the product or its suitability for a special purpose. Responsibility for proper and correct use of the product belongs to the user. Regarding any damage within the warranty period, SeDenta Health Services Co. Ltd.’s responsibility is only to repair or replace the relevant product.
Limitation of Liability: Regarding the use of this product, SeDenta Health Services Co. Ltd. is not responsible for any direct, indirect, special, or any kind of damages.
KEEP OUT OF THE REACH OF CHILDREN. FOR DENTIST USE ONLY.
It was produced in Germany for SeDenta Health Services Co. Ltd.